SMTL Dressings Datacard

Product Name: MELOLIN

Classification Name: Dressing Perforated Film Absorbent BP (Type 1)

Manufacturer: Smith & Nephew Medical Ltd

Description

Melolin consists of a film of poly(ethylene terephthalate), onto which is bonded an absorbent layer consisting of a mixture of cotton and polyacrylonitrile fibres, backed with a layer of an apertured non-woven cellulose fabric. The plastic film is present to prevent the dressing adhering to the surface of the wound, and is perforated to allow the passage of exudate from the wound into the body of the pad.

Indications

Melolin may be used on its own to dress dry sutured wounds, superficial cuts and abrasions[*** linked me-ab-00 ***], and other lightly exuding lesions. It may also be used as the primary wound contact layer for more heavily exuding wounds, if backed by a second absorbent dressing.

Contra-Indications

In common with other perforated plastic film dressings, Melolin should be used with caution in the treatment of leg ulcers that produce copious quantities of very viscous exudate. Under these circumstances, the exudate may become trapped under the dressing, leading to maceration and inflammation of the surrounding skin.

Method of use

Melolin has a relatively low absorbent capacity and is therefore intended for use on lightly exuding wounds. In these situations, a single dressing may simply be secured in position with surgical tape. On more heavily exuding wounds, Melolin may be used as a wound contact layer, beneath a secondary absorbent pad held in position with tape or a bandage.

Frequency of Change

The frequency with which the dressing should be changed depends entirely upon the nature and condition of the wound.

Presentation

[*** linked ps-melolin ***]Pack Shot Melolin is available in two forms: individually wrapped in a paper peel pouch, and steam sterilised; or in bulk, non-sterile.

Sizes

Individually wrapped 5 cm × 5 cm* 10 cm × 10 cm* 20 cm × 10 cm* Bulk 5 cm × 5 cm 10 cm × 10 cm 20 cm × 10 cm 20 cm × 30 cm 50 cm × 7 m * available on Drug Tariff.

Revision Details

Revision Author Dr S. Thomas Revision No 1.4 Revision date 1997/12/16

Please inform SMTL of any inaccuracies or changes required in this datacard.

Disclaimer

This datacard has been prepared as an aid to contracting for purchasers in the NHS in Wales, and should not be used as a substitute for literature, instructions or training from the manufacturers of the devices.