SMTL Dressings Datacard

Product Name: Granugel

Classification Name: Dressing Hydrogel, Amorphous

Manufacturer: ConvaTec Ltd

Description

Granugel is a sterile gel containing pectin and sodium carboxymethylcellulose in a clear, viscous vehicle containing propylene glycol. The dressing has the ability to absorb excess exudate from exuding wounds but donate moisture to dry necrotic tissue or slough.

Indications

Granugel is indicated for the management of partial and full thickness wounds including including leg ulcers, pressure sores and surgical wounds. In the management of necrotic and sloughy wounds, Granugel facilitates cleansing by donating moisture to the dead tissue causing it to become rehydrated, thus facilitating autolytic debridement.

In the management of granulating wounds, Granugel maintains a moist environment which is claimed to facilitate the formation of granulation tissue.

Contra-indications

Known sensitivity to the gel or its components.

Method of use

Granugel should be applied directly into the wound up to the level of the surrounding skin and covered with a moisture retentive dressing e.g., Granuflex.

Frequency of Change

Granugel should be changed when the cover dressing leaks or for routine dressing changes. Frequency of dressing change will depend on the state of the wound and level of exudate. On necrotic and sloughy wounds Granugel can be left on for up to 3 days. For clean granulating wounds Granugel can be left on for up to 7 days. The gel can be rinsed out with sterile saline.

Warnings

When a wound is clinically infected Granugel may be used together with appropriate antibiotic therapy and the progress of the wound monitored carefully under medical supervision. In the presence of anaerobic infection occlusive therapy is not recommended.

Presentation

Granugel is available in single use tubes, sterilised by autoclave.

Storage

Granugel should be stored in temperatures below 25\(deC, avoid freezing.

Sizes

15g tube* *Available on Drug tariff.

Bibliography

1. Thomas S, and Hay NP, Fluid handling properties of hydrogel dressings, Ostomy and Wound Management, 1995, 41, (3), 54-59. 2. Hofman D, The healing of cavity wounds: Preliminary use of Granugel in pressure sores, Nursing Times, 1996, 92, (29), 64-68.

Revision Details

Revision Author Dr S. Thomas Revision No 1.5 Revision date 1997/12/16

This datacard has been prepared from data provided by the manufacturer and/or from published literature.

All materials copyright © 1992-2002 by the Surgical Materials Testing Lab. unless otherwise stated.

| Home | Alphabetically | SMTL | World Wide Wounds | Disclaimer |

http://test-dressings.smtl.co.uk/Dressings/granugel.html Last Modified: Thursday, 02-May-2002 13:56:44 BST