SMTL Dressings Datacard

Product Name: COMFEEL: PLUS TRANSPARENT DRESSING

Classification Name: Dressing Hydrocolloid, Transparent

Manufacturer: Coloplast Ltd

Description

[*** display ps-comfeel.trans-01 ***]

[*** linked comfeel.trans-01 ***] Comfeel: Plus Transparent Dressing consists of a semipermeable polyurethane film coated with a thin layer of a self adhesive mass containing sodium carboxy-methylcellulose (NaCMC). The dressing is permeable to water vapour but impermeable to exudate and micro-organisms. It is waterproof, allowing the patient to bathe or shower.

Indications

Comfeel Plus Transparent Dressing has very limited absorbency and therefore should be used in the treatment of lightly exuding superficial wounds including [*** linked burn-superf-01 ***] minor burns, [*** linked abrasion-01 ***] abrasions, and [*** linked donor-site-01 ***] donor sites after haemostasis has been achieved.

The dressing may also be used prophylactically to help prevent friction damage to the skin of patients who are at risk of developing [*** linked p-sore-grade1-01 ***] pressure sores.

Contra-Indications

Comfeel: Plus Transparent Dressing should not be applied to patients who are known to be sensitive to one of the ingredients.

Method of use

An appropriately sized dressing is partially removed from its paper backing using the tab provided, and placed over the wound using a non-touch technique. In order to ensure good adhesion to the surrounding skin, a minimum overlap of 1.5 cm from the margin of the wound should be allowed.

Frequency of Change

On clean epithelialising wounds, the dressing should be left in place for as long as possible. On chronic wounds or those containing small amounts of slough, the dressing should be changed more frequently.

Warnings

Comfeel: Plus Transparent Dressing should not be applied to infected, or heavily exuding wounds or those containing significant amounts of slough or necrotic tissue.

Presentation

Comfeel Plus Transparent Dressing is presented individually wrapped in a peel pouch, sterilised by Beta irradiation.

Sizes

5.0 cm × 7.0 cm 9.0 cm × 14.0 cm 15.0 cm × 20.0 cm

Revision Details

Revision Author Dr S. Thomas Revision No 1.5 Revision date 1997/12/16

Please inform SMTL of any inaccuracies or changes required in this datacard.

Disclaimer

This datacard has been prepared as an aid to contracting for purchasers in the NHS in Wales, and should not be used as a substitute for literature, instructions or training from the manufacturers of the devices.